Project

# Build Your Own Low-Resistance Meter

October 26, 2016 by Robin Mitchell

## This project will show you how to make a device that can measure resistances as low as 0.1 ohms.

Learn how to build your own low-resistance meter!

You probably already have a DMM for measuring resistance, but can it be used with resistances below 1Ω? And if so, are those low-ohm readings reliable?

This project will show you how to make your own low-resistance meter; it uses only a handful of components and can measure resistances as low as 0.1Ω.

### The Theory

Measuring resistances can be done using a variety of methods (Wheatstone bridge, RC calculation), but in this project, the chosen method is to use the most fundamental equation in electronics:

$$V = IR$$

A constant-current source will establish the current through the resistance under test and measure the voltage drop that the resistance produces. This voltage drop will then be amplified and fed into a standard multimeter. The magnitude of the voltage will be equal to the resistance in ohms (e.g., 1V = 1Ω). We will need to select a current that produces reasonable voltages for the amplifier stages that follow the constant-current stage, and we can do this by using the above equation and inserting expected values for R (i.e., less than a few ohms).

One important consideration is the op-amp's input offset voltage, which is modeled as a voltage source in series with the op-amp's inverting or noninverting input terminal. This voltage is  multiplied by the op-amp's noninverting gain, and it is a source of error because it can make the output voltage lower or higher than what we would expect from an ideal circuit. So we want to design our circuit such that the effect of this offset voltage will be minor. If your op-amp has offset-null functionality, you can use that to reduce the amplitude of the offset voltage, but we're using the LM358, which does not include offset-null pins. Instead, we can easily reduce the influence of the offset voltage by ensuring that the signal of interest is much larger than the offset voltage, which is ±2mV for the LM358.

Our goal is to measure resistance as low as 0.1Ω. This means that we have to choose a constant-current source that creates a voltage significantly larger than 2mV when the current is passing through a resistance of 0.1Ω. This is a trade-off, because higher currents have disadvantages and lower currents reduce the voltage drop across the resistance under test. The problems with higher current are the following:

• Higher power consumption, whereas lower power consumption helps with portability.
• Lower currents result in less heat generated by the constant current source circuit.
• Lower currents reduce the power dissipation and hence temperature increase of the resistance under test; with lower current, we can measure the resistance of circuit elements that are more susceptible to heat damage (thin wires, for example).

The current chosen for this circuit is 100 mA. This amount of current is not too high yet it generates 10mV across a 0.1Ω resistor, and 10 mV is adequate considering our ±2mV offset voltage.

The constant-current source consists of

• U1A – LM358
• Q1 – 2N3055 (TO-3 package)
• RV1 – potentiometer for adjusting the reference voltage applied to the op-amp's noninverting terminal
• R1 & R2 – divider (1V from RV1 corresponds to constant current of 100mA)
• R3 – sense resistor (1Ω, 1W, metal film, 1% tolerance)
• P2 – two terminals for connecting the resistance to be measured

With a constant current of 100mA through the 1Ω sense resistor, the power dissipation is 0.1W (hence the choice of 1W). Q1 will conduct 100mA as long as a resistance is connected to P2, and I chose the TO-3 package to ensure that the transistor does not overheat. The specific part used for Q1 is not that important as long as the transistor can handle 100mA of collector current and is NPN.

The next stage after the constant-current source is a differential amplifier with a gain of 1 and an offset voltage adjust. We are using a "differential" amplifier here because we want to detect the voltage drop across the resistance under test, i.e., the difference between the voltage on one side of the resistance and the voltage on the other side of the resistance.

The differential amplifier consists of

• U1B – the op-amp
• R4, R5, R6, and R7 – these resistors configure U1B as a differential amplifier
• R8, R9, and RV2 – offset adjust

The circuit consisting of R8, R9, and RV2 allows us to add an adjustable offset voltage to the output of the differential amplifier. This feature can be used to compensate for the op-amp's input offset voltage or other error sources. Please refer to the calibration section (below) for details on how to implement this compensation circuit.

The last stage is an amplifier with gain of 10. This additional gain sets the overall measurement ratio to the convenient value of 1:1, i.e., 1Ω of resistance produces 1V at the output.

• U2A, RV3, and R10 – noninverting amplifier with gain of 10 (RV3 set to 90K)
• U2B – output buffer

### BOM (Bill of Materials)

 Component Ident Value U1, U2 LM358 – DIP 8 R1, R4, R5, R6, R7 100K resistor R2, R10 10K resistor R3, R8 1R 1W metal film resistor, 1% tolerance RV1, RV2, RV3 100K linear potentiometer Q1 2N3055 BJT, TO-3 C1, C2 100nF decoupling capacitors

### Construction

How the circuit is constructed is up to you but here are a few ideas:

• Project box – Use an internal 9V battery and external connectors to keep the circuit in one small box.
• Multimeter attachment – Using a few banana plugs, you could create a circuit that fits directly onto a multi-meter.
• A meter – Going all the way, you could purchase a small voltage meter and house the whole project in its own package to make your own measurement device!

##### The low-resistance meter as a breadboard circuit.

The photo above shows the three potentiometers:

• the one on the left controls the constant-current source
• the one in the middle controls the differential amplifier offset
• the one on the right controls the gain of the output stage

The red, green, and black wires leaving the breadboard are for +5V, 0V, and -5V, respectively. The brown and red wires that go off toward the top of the image are for the resistance under test, and the green wire going to the right is for connecting the low-resistance meter's output to the input of a multimeter.

Note: You will need to ensure that the multimeter's common input is connected to the ground of the low-resistance meter.

### Power

This circuit requires a split power supply for full functionality. However, note that the negative rail is used only in the circuit that adds an adjustable offset voltage to the output of the differential amplifier. If you can get adequate performance without this compensation circuit, the negative rail is not needed. If you're not using the LM358, keep in mind that the input common-mode voltage range of your op-amp must extend almost all the way to 0V, because we are dealing with input voltages near 100 mV.

The power-supply requirements are rather flexible (but don't exceed your op-amp's maximum supply voltage). You need to make sure that the power supply can source sufficient current (at least 200mA, considering that the current source alone requires 100mA). Also, be aware that the power dissipation of Q1 is proportional to the positive supply voltage, so keeping the input voltage as low as possible will reduce Q1's power dissipation.

I recommend ±5V supplies; for the negative rail you can use a Negative Voltage Generator.

### Calibration

The first part of the circuit to calibrate is the constant-current source. The easiest method is to use a multimeter (connected to P2) to measure the constant current.

Adjust the value of RV1 until the measured current is 100mA. Start with RV1 adjusted to its minimum resistance. This minimizes the initial constant-current setting and thus prevents potentially harmful amounts of current through Q1 and R3; also, the resulting power dissipation will lead to increased component temperatures, and hot transistors can cause serious contact burns.

With the constant current set, we need to compensate for error in the output of the differential amplifier. You can do this by measuring a known resistance and then adjusting RV2 until the output of the differential amplifier corresponds to the known resistance (for example, a resistance of 1Ω should produce a differential output of 100mV), or you can measure the voltage across a small resistance using a precise voltmeter and then adjust RV2 such that the differential amplifier's output is equal to the measured voltage.

The final calibration step is to adjust RV3 so that the gain of the U2A amplifier is equal to 10. Measure the noninverting input voltage of U2A and adjust RV3 until the output is 10 times the value of the input.

### Summary

With the circuit complete, you can now test your circuit to see if it is functioning correctly. If all goes well, you should now have a low-resistance meter that works in conjunction with an accurate multimeter.

So where could this be used? Personally, I built this circuit so that I could measure the resistance of mains wires (when dead, of course!) for some electrician practice. Instead of purchasing a very expensive electrical kit (costing at least $500), this circuit allows me to practice at a price of a few bucks. This circuit can be used with a pair of small pogo pins to probe small PCB traces. It can also be used to measure contact resistance (which can sometimes cause issues in circuits that rely on mechanical contact). Give this project a try for yourself! Get the BOM. ### You May Also Like 20 Comments • Bverysharp October 28, 2016 Looks great! I may build this! Like. • Robin Mitchell October 28, 2016 Please do! It would be better to use op-amps that have better characteristics than the LM358 but if on a budget can be OK! Like. • frpr666 October 29, 2016 I built Dave Johson’s 1A current injector. (http://www.discovercircuits.com/DJ-Circuits/1ampcurrent.htm). I’m happy with it. CONS: not difficult, portable, allows 4 terminal measurement. Interesting point about the design is the usage of MOSFET and comparator LM393 as constant 1A source. Like. • Barbo November 11, 2016 2N3055 for 100mA current only?....why?......It needs a better explanation..A BC337, BC338 can be used loosely. Like. • Robin Mitchell November 21, 2016 Explanation : 100mA through transistors such as a BC548 and 2N3904 cook! I wanted to use a cheap transistor that was common place and had a decent heat dissipation capability hence the 2N3055 Like. • P Phil-S November 15, 2016 Low reistance measurements are essential for checking out domestic wiring (yes, with the power off) for continuity and making sure the overall resistance is low enough to trip the MCB’s and other protection devices - the lower the conductor resistance, the faster and more reliably the trip will go - could be a life-saver. Nicely done. Like. • M magicChristian November 21, 2016 One point i do not agree with Robin: The power of Resistance R3 with 100 mA and 100 mV is only 0,01 W. And it is always constant as long as the system works, therfore one can use a 0.25 W component with good stability. To measure real low resistance we have to use 4 wire connection. This can be done having P2 with 4 pins, 2 for the current and serate 2 for measuring the voltage. Like. • Robin Mitchell November 21, 2016 Good call on the 4 pin measurement. This project however is just a simple circuit that can be used for low ohm resistance measurements (a weekend project say). Its more important for R3 to be a high precision resistor as opposed to being a 1W resistor but those are what I had freely available and are cheap! Like. • S shadow3 December 13, 2016 Sorry I am confused… Why do we even need the Vee and gnd connection in the two potentiometers? Don’t me just need the two terminals when using it? Like. • S Saif3011 February 27, 2017 I need to build a low resistance measurement but I need to use lock in amplifier. I need to generate frequency by using MC14060 which the frequency must below 1kHZ and not harmonic of 50Hz. Do you have any idea how to realise it ? Like. • M mohammed1984 February 27, 2017 hi this is my first sign in.. thank u for your great job.. Like. • HARSHA KILLER March 22, 2017 i want the full image in detalied about The low-resistance meter as a breadboard circuit. because i am just 1st btech student. i am a cse student Like. • N npublici April 12, 2017 My interest in building this circuit is to measure the primary of automotive single cylinder spark coils.Most are below ten ohms. Will this device have enough range? Also what would “better’ Op amps than the LM358 be for this application? Like. • Polymorph May 23, 2017 0.1 Ohm? Did you mean 0.1mOhm? At 1V per ohm output and 2V scale on a 3.5 digit meter, that should get down to 1mV = 1mOhm. If it autoranges down to 200mV scale, that would give you 0.1mOhm minimum measurement. Otherwise, everything between the constant current source/trace under test and the voltmeter is wasted effort. 100mA through 1mOhm = 0.1mV, which means 1mOhm resolution on the 200mV scale. Like. • Polymorph May 23, 2017 For measurements this low, I consider 4 point measurements essential. Otherwise, you are also measuring the contact resistance of the probes to the trace or wires under test, which can easily swamp the desired measure. Kelvin probe clips are about$20 on a certain online auction site. Or you can make some dual tip probes using spring aka pogo terminals.

Like.
• HEC-2016 June 28, 2017

Polymorph and others, You are correct about the 4 point method, but there are so many crude components here that I don’t think it would matter much.  Just tie your test leads together and tune out the offset.

My approach on this would be to use a low input offset current/voltage, rail to rail Opamp, such as the TLC2274, and go with a voltage reference and precision resistors to eliminate the pots.  You could also use an N-Channel MOSFET with a 1amp rating to replace the overkill BJT.  For the 4 wire setup, just make your high current connections one set of big leads, or use a speaker terminal connector panel mounted on your box.  Then have a small set of sense leads that you would clip onto the load as close as possible to it’s lead to body interface on one end, and the input to the Diff-amp on the other.  A common-mode choke and/or pi filter in between the two wouldn’t be a bad idea.  By precision, I mean .1% of each other, as in the same package of 4 identical Rs, each the same resistance.  The average R is far less important than the matching Rs on Diff-amps, so .5% to 1% F/S on the absolute resistance is fine if they have the same tempco.  Hand pick ‘em if you have to.  That 1 ohm / 1W resistor is good for low R measurement, but I would go with .1% and not 1%.

Guys, if you’re going to make something like this, do it right and do it once.  You will then have an instrument that you can trust for life.  Put it in a nice metal enclosure, with good connectors, power switch and “on” LED, A/C adapter input, and a tidy, clean PCB.  If you can’t do a custom one, then use Perfboard and use a socket for when you fry the Quad.  A fuse between the collector and VCC wouldn’t be a bad idea.  Fast blow, 1A. Heatsink that BJT or MOSFET with galvanic isolation.

I’ve made variations on this type of circuit that use high voltage transistors to put current through high Mega-Ohm resistors for the same purpose.  Stay below 100v if you do this, otherwise you will need to coat things in corona dope for long term service.  HV has a way of making its own “resistors” on anything porous and exposed to the atmosphere.  Never use breadboards for HV.

BTW, If you put another Diff-Amp at the front end (again with precision resistors), you’ll have a nice voltage to current converter.

Like.
• R
rcrypto August 29, 2017

I have 2 question regarding this circuit! Firstly, wouldn’t you think a simple differential pair with an amplifier for gain could also work. Secondly, how could we make the pot to adjust itself with respect to the threshold we fixed.
Would this circuit work for resistance that’s large with a small change, for example, if i want to detect the changes in the resistance as low as 100 milli ohm with its absolute resistance of say 300 ohms ?

@robin mitchell
can you please post your email id so i can contact you for further details, I am currently working on a project that I may need assistance with.

Like.
• S
swapnil14327 February 17, 2020

I need a similar circuit but my driving voltage is 3.3V via battery.
I need to measure temperature via resistor sensor RTD, NTC, PTC etc.
as my application is battery powered i need very low power circuit.