In this project, we'll build a weighing scale using a C8051 microcontroller development kit, a load cell, and the Simplicity Studio IDE. The measured weight will be displayed on an LCD screen.

### LCD Display

The LCD used in the project is part of the Silicon Labs CP2400DK development kit.

##### Figure 1. Silicon Labs CP2400DK development kit (C8051 microcontroller and LCD controller). Image courtesy of Digi-Key.

The manufacturer part number of the LCD device itself (figure 2) is VIM-878-DP-RC-S-LV. If you include the decimal point and the apostrophe (for each digit), this "14-segment" LCD display becomes a 16-segment display. And when summing all the 16-segments of the eight digits together, we see a total of 128 segments. Such a 128-segment display mandates the use of a 128-segment driver, and Silicon Labs has decided to use driver p/n CP2400 (Figure 3).

##### Figure 3. 128-segment LCD driver. Images courtesy of Silicon Labs (pages 1 and 22).

The load cell that I chose to use is made by Uxcell, model # a14071900ux0057.

##### Figure 5. The load cell's Wheatstone bridge strain gauge schematic.

For this project I'll be applying 5.0VDC as the excitation; therefore, when a 10kg load is applied to the load cell, its VOUT will be 5.0mV (±0.75mV). And because 5.0mV is such a small voltage, it will need to be amplified prior to being sent to the microcontroller.

Side note: Some microcontrollers—not the one used in this project—have differential ADC inputs; single-ended inputs are considered standard for most microcontrollers. When using a microcontroller with a single-ended ADC input and when the sensor in question provides a differential output signal, a differential amplifier must be utilized. This type of amplifier converts a differential signal into a single-ended signal that can be measured using a standard single-ended ADC.

### Instrumentation Amplifier

The differential amplifier that I chose to use is actually called an instrumentation amplifier, or in-amp. I began this project using the INA126PA in-amp from Texas Instruments—I purchased this device from Digi-Key for $3.15. However, when it failed to perform as advertised and after troubleshooting it for about an hour, I decided to move to Plan B: use the AD627 (from Analog Devices), which I had also purchased. I bought this part—quantity one—from Digi-Key for a whopping price of$8.45! I had no idea that these in-amps cost so much money! Fortunately, this device worked flawlessly. Hence, I think I'll be using in-amp devices from Analog Devices in the future as opposed to TI parts, although generally speaking, I'm a fan of parts, and prices, from TI. Setting the gain for both of these in-amps is simple: all that’s required is one external resistor. Also, both the TI and the Analog Devices part are advertised as single- or dual-supply devices, and I much prefer to use a single-supply.

##### Figure 6. Analog Devices instrumentation amplifier (AD627) simplified schematic. Image courtesy of Analog Devices (page 14).

Item # Description / Source Cost (each) Other Information
1  C2400DK development kit $148.75 User Guide Quick-Start Guide C8051F930 datasheet Note: schematics are on pages 25-31 of the User Guide. 2 Breadboard$8.98  or equivalent
3  Jumper wire kit $6.20 or equivalent 4 10kg load cell$8.14  or equivalent
5  Instrumentation amplifier $8.45 Datasheet 6 1 kΩ potentiometer$2.41  Datasheet
7  Machine Screws (M4-0.7 x 25mm).
Qty 2
$0.76 or equivalent 8 Machine Screws (M5-0.8 x 25mm). Qty 2$0.76  or equivalent

### Making the Connections / Schematic

The microcontroller uses the SPI interface (SPI channel 1) for communicating to the LCD driver—figure 7 below shows Simplicity Studio's hardware configuration GUI for how the C8051's SPI is configured.

#### Choosing the Gain for the Instrumentation Amplifier

As mentioned before, the excitation voltage applied to the load cell is 5.0VDC, which will produce an output voltage of 5mV (at full load of 22 lbs). Though the microcontroller's ADC has an input range of 0 to 3.30VDC, I want 3.00VDC to correspond to the full load (22 lbs). This will provide some headroom for the load cell's overload, which is 120% of full load, or 26 lbs. Given these conditions, I can calculate the necessary gain for the in-amp, and then choose the appropriate value of the gain resistor (RG).

$$5mV*Gain=3000mV$$

$$Gain = \frac{3000mV}{5mV}=600$$

RG equation (per AN627 datasheet page 22):

$$R_G = \frac{200k \Omega}{Gain-5}$$

$$R_G = 336 \Omega$$

### Configuring the Microcontroller Development Kit

Prior to powering up the microcontroller development kit, after all the connections have been made, be sure to configure it as follows:

Jumpers:

• J11: VBAT to WALL_PWR
• J12: VDD to VIO
• J17: VBAT_PIN to VBAT

Switches:

• SW4: set to "2 CELL"
• Power switch (SW5) to "OFF" position

Cables:

• Connect the ribbon cable debug adapter to J9
• Apply 5.0VDC to connector P2.

### Attaching the Load Cell Base Plates

The load cell itself has holes with, to my surprise, screw threads. This makes it very handy when attaching the load cell to base plates. Again, to my surprise, one set of holes is larger than the other. Hmmm, I'm not sure why the manufacturer did this but, nonetheless, I was able to find the appropriate screws at the local hardware store.

• Quantity two: machine screws, M4-0.7 x 25mm
• Quantity two: machine screws, M5-0.8 x 25mm
• Note: the length of these four screws (25mm) is dependent upon the thickness of the spacers and of the base plates themselves.

### The Firmware

Similar to another project of mine (Adding an LCD and Keypad to a Tachometer and Speedometer), for this project I heavily leveraged Silicon Lab's example LCD project (CP240x_LCD_Example), which made my firmware-writing task much easier.

The output of the in-amp is connected to port 0.6, which is the input to the microcontroller's ADC. Below is a list of features/goodies that I've incorporated into the firmware:

• The firmware takes and averages 200 samples of the measured voltage prior to sending the data to the LCD screen.
• If the applied load is greater than or equal to 23 lbs, the LCD will display "OVERLOAD."
• When the system is first energized, the firmware will display the tare weight (i.e., the weight at startup becomes the zero weight).
• The LCD is configured to display one digit following the decimal point.

                    //----------------------------------
// Main Application Loop
//----------------------------------
while (1)
{

//-----------------------------------------------------------------------
// Measure Analog Value
//-----------------------------------------------------------------------
//
//

// Initiate a Conversion
AD0WINT = 0;                        // clear window detector flag
AD0BUSY = 1;                        // initiate conversion

// Wait for conversion to complete

//                           Vref (mV)
//   measurement (mV) =   --------------- * result (bits)
//                       (2^10)-1 (bits)

mV *= VREF;
mV /= 1023;
lbs = (float) (mV * 22.0 / 300.0); // The multiplying factor is actually 22lbs/3000mV, but 300
// is used for displaying one digit after the decimal point.
if(zero_scale == 0)		    // Used for zeroing the scale during power-up.
{
zero_offset = lbs;
zero_scale = 1;
}
lbs = (lbs - zero_offset);
AverageAccumulator += lbs;	    // Add the current lbs measurement to the accumulator.
AverageMeasurements--;		    // Decrement the measurement counter.
if(AverageMeasurements == 0)
{
// Calculate the average value: divide the summed AverageAccumulator by the
// number of measurements.
lbsAverage = (AverageAccumulator / 200.0);

AverageAccumulator = 0;	   // Reset
AverageMeasurements = 200; // Reset
if(lbsAverage >= 230)	   // If the measured weight is greater than equal to
// 23.0 lbs, then display "OVERLOAD" on the LCD screen.
// The load cell is rated at 10kg (or 22 lbs), with
// safe overload of 120%, or 12kg (26 lbs).
{
}
else sprintf(display_string, "d lbs ", (unsigned int) lbsAverage);
}

//-----------------------------------------------------------------------
// Update LCD
//-----------------------------------------------------------------------
//
// Update the LCD Display
//
LCD_OutString(display_string);
}


### Building and Loading the Code, and Verifying the Accuracy of the Scale

After downloading, building, and loading the code, I used my kitchen scale as a reference/comparison for taking weight measurements. As seen in the video (below), I place various weight samples (my old textbooks from college!) first on the kitchen scale and then on the load cell system. And as you can observe, both scales are extremely close in their displayed measurements.

### Next Steps for Making a Real Product

If you, or I, decide to make an actual weight-measuring system based on this project using a customized PCB design, be sure to follow the grounding and layout recommendations as described in the datasheet (page 20). Also, replacing the wooden base plates with metal plates would increase the stability and robustness of the physical design.

Happy weighing!

Give this project a try for yourself! Get the BOM.

• catalin_cluj 2017-10-13

I think it’s wrong of you to say something was wrong with the TI part without clearly stating what it was, or that it could have been the way you used it.
I have no connection with TI but such vague statements and conclusions are very wrong.

• RK37 2017-10-16

You’re right, we need to be careful about stating that parts are defective. We revised that section and will leave the revised version in place until the author has clear proof that the part was defective.

• Kevin Wolfe 2017-10-27

How can you write an article about load cells and not mention the HX711?    They can be had on eBay soldered onto breakout boards for about $0.75 each - shipped! NodeMCU even has an HX711 library that can read out the values from an ESP8266 via LUA script. Check out https://nodemcu-build.com • Nick Davis 2017-10-27 Hi Kevin. Thanks for sharing this information…I hadn’t heard about the HX711 before now. I’ll definitely consider using it on any future scale projects. • John Leek 2017-11-01 I have used the INA126P a lot and never had any problems with this part - So why have you got problems ? it must be you can not read the manufactures data sheet correctly and or you connected it wrongly. Please return to this device and find out what you did wrong and correct it you may learn something. My students all use this part and it always works.. • Privatier 2017-11-01 Are you trying to sell us an expensive development system? How about an Arduino and a cheap LCD display, together for less than$5? How do you think can anybody sell an electronic scale for less than \$20?

• JamRacing 2018-03-04

Hello.I have used a picaxe 08m2 connected to INA 125 for my load cell project.Please explain how to calibrate.